for standard waveforms. The Agilent Technologies A Use the A to generate complex . Manual language options (please specify one). ABA US. For Safety information, Warranties, and Regulatory information, see the last page in this manual. Agilent A. Option Phase-Lock Assembly. oscilloscope or DMM and send it to your HP A for output. 3-year warranty. With your HP A, you get operating and service manuals, a quick reference .
|Country:||Republic of Macedonia|
|Published (Last):||21 May 2007|
|PDF File Size:||11.96 Mb|
|ePub File Size:||15.98 Mb|
|Price:||Free* [*Free Regsitration Required]|
Chapter 1 Quick Start To set a dc offset voltage To set a dc offset voltage At power-on, the function generator outputs a sine wave with a dc offset voltage of 0 volts into a 50 W termination. Chapter 3 Features and Functions Output Configuration Instrument State Storage You can store up to three different instrument states in non-volatile memory.
Decibels are a ratio or comparison unit and have no absolute meaning without knowledge of a reference or comparison unit. The stop frequency is automatically adjusted if you select a function whose maximum frequency is less than that of the currently active function.
You must specify the source from which the function generator will accept a trigger. For example, the following command string guarantees that the first trigger is accepted and executed before the second trigger is recognized.
Select a Web Site
Chapter 3 Features and Functions System-Related Operations Display Control To speed up the rate at which the function generator can make configuration changes, or for security reasons, you may want to turn off the front-panel display.
To Set A Dc Offset Voltage Chapter 1 Quick Start To set a dc offset voltage To set a dc offset voltage At power-on, the function generator outputs a sine wave with a dc offset voltage of 0 volts into a 50 W termination. If you have questions relating to the operation of the A, call in the United States, or contact your nearest Agilent Technologies Sales Office.
Page 30 Chapter 1 Quick Start To rack mount the function generator To rack mount a single instrument, order adapter kit Don’t show me this message again. Learn More – opens in a new window or tab. Then, rotate the handle to the desired position.
Agilent a output Disable – Discussion Forums – National Instruments
There are 1 items available. Subject to credit approval.
The following steps show you how to change the amplitude to 50 mVrms. Duty cycle is specified as a percentage and represents the amount of time per cycle that the square wave is high. This section is only an SCPI users, see page To Set The Output Frequency Chapter 1 Quick Start To set the output frequency To set the output frequency At power-on, the function generator outputs a sine wave at 1 kHz with an manuaal of mV peak-to-peak into a 50 W termination.
Internal modulation waveshapes range in length from 2, points to 4, points. Originally Posted by weeyndha. See other items More Used only in the triggered burst mode internal or external source. Michael Grieger 28 Jul How reliable is it?
Back to home page. If you continued change to a function that is not allowed with the selected modulation, the modulation mode is turned off. As noted in the first post, the vendor says the option is not available. If anything is missing, contact your nearest Agilent Technologies Sales Office. I can go for that if i’m sure that it does not cause any other effect. Other MathWorks country sites are not optimized for visits from your location.
Agilent Technologies Sales ance of a U. System-related Operations Chapter 3 Features and Functions System-Related Operations System-Related Operations This section gives information on topics such as power-down recall, self-test, error conditions, and front-panel display control.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
Input port and input output port declaration in top module 2. Creating Arbitrary Waveforms 16, points long and the full range from 0 to 4, codes. The function generator uses line to send a hold-off signal. The status registers, the error queue, and all GPIB configuration states are left unchanged when a device clear message is received. Page 18 – If the function generator does not turn To rack mount two instruments side-by-side, order lock-link kit and flange kit Power from the fundamental frequency, and its harmonics, is transferred to spectral components of the rectangular sampling function.
Chapter 2 Front-Panel Menu Operation A front-panel menu tutorial Menu Example 2 The following example shows how to use the recall menu feature as a shortcut to set the command back to its original setting. To print the manual completely, please, download it.