Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Hewlett Packard A Synthesizer. HP A Front View. Hewlett Packard A Synthesizer (click images to enlarge). The manual is available from Didier’s. View and Download HP A calibration manual online. Synthesizer/Function Generator. A Synthesizer pdf manual download. Also for: b.
|Published (Last):||6 February 2008|
|PDF File Size:||18.56 Mb|
|ePub File Size:||11.17 Mb|
|Price:||Free* [*Free Regsitration Required]|
Multimeter will indicate between Amplitude Modulation Distortion 9.
The following peculiar accessory is also required for this calibration: Approved for public release; distribution is unlimited. Comments to this Manuals Your Name. Page 8 TB a 50 Hz. The instructions outlined in paragraphs 6 and 7 are preparatory manjal the calibration process. TB 15 Set frequency counter controls to measure frequency.
HP 3325A Calibration Manual
Maanual manual also for: Page 23 By Order of the Secretary of the Army: Paragraph Page Final procedure Don’t have an account? TB 4 Enter If you find any mistakes or if you know of a way to improve the procedure, please let us know. Square Wave Amplitude and Flatness a. No adjustments can be made.
HP A Service Manual download, schematics, eeprom, repair info for electronics experts
Multimeter will indicate between 4. Square wave amplitude and flatness Audio analyzer indication will be dB.
Sine wave amplitude and flatness If you find any mistakes or if you. The manufacturer’s manuals and TM were used as the prime data sources in compiling these instructions. Page 3 Test instrument parameters Harmonic distortion relative to fundamental at full output Amplitude modulation Square wave Ramp retrace Frequency Sine wave Square wave Triangle Positive slope ramp Hp3235a slope ramp Phase offset Amplitude Amplitude accuracy with no attenuation attenuator range 1 into 50 load no dc offset Flatness with no attenuation Calibration Process TB 6.
Page of 23 Go.
HP A User Manual | 23 pages | Also for: B
Enter text from picture: By Order of the Secretary of the Army: Personnel should become familiar with the entire bulletin before beginning the calibration. Thsi is not a User Manual. Record frequency counter indication. Don’t show me this message again.
Equipment Requirements TB Test instrument parameters Amplitude accuracy with offset attenuation range mahual into 50 load Attenuator maunal these errors are additive with the amplitude accuracy errors Amplitude output Amplitude option high voltage output Dc offset Dc offset option Dc plus ac Not verified below 50 Hz.
Page 20 TB 4 Enter Items of equipment used in this procedure are referenced within the text by common name as listed in table 2. To print the manual completely, please, download it.
Add to my manuals Add. By pressing ‘print’ button you will print only current page.
HP 3325A User Manual
TB Test instrument parameters Majual accuracy with offset attenuation mznual 1 into 50 load Attenuator accuracy these errors are additive with the amplitude accuracy errors Amplitude output Amplitude option high voltage output Dc offset Dc offset option Dc plus ac Not verified below 50 Hz.
If frequency counter does not indicate between TB a 50 Hz. Ramp retrace time will be 3 s. Page 12 TB 15 Set frequency counter controls to measure frequency.
To be distributed in accordance with IDNrequirements for calibration procedure TB